Shrinkage Template Scale TF178
Shrinkage Template Scale, suitable for all fabric shrink testing standards. Shrinkage Template Scale complies with testing methods like AATCC 135/150/160/179, BS EN 3759, ISO 3759, etc.
Shrinkage Template & Ruler is used to mark the origin size of the testing fabric and track its shrinkage status by the scale on the ruler, built with high-transparent acrylic makes this template ruler very easy to observe the dimension change and calculate the shrinkage scale and stretch scale. And it combines the different test standards in one template, including 500mm, 350mm, 250mm. Frame size is 610mm x 610mm; Shrinkage Scale is 0%~20%, and Stretch Scale is 0%~ 20%.
- Dimension of template 250mm, 350mm, 500mm
- Shrinkage Scale 0%~20%
- Stretch Scale 0%~20%
|Length:||650 mm||Width:||650 mm||Height:||50 mm|
|AATCC 135||AATCC 150||AATCC 160||AATCC 179|
|BS EN 3759||ISO 3759||BS 4931|
What is Fabric Shrinkage?
Fabric shrinkage is a common phenomenon that the defect of the fabric which is subjected to home laundering procedures. To determine the quality of the fabric, shrinkage is an important element which is needed to consider. There are many elements to affect the dimensional changes of the fabrics such as the washing temperature, rinse temperature, dying procedures, etc.
Scope and Purpose of Fabrics Shrinkage Test
The shrinkage rate of textiles is an important item which must be tested in the textile industry of every country at present. At present, the countries with different standards of shrinkage rate of textiles carry out their own standards.
This testing method is suitable for the size change of woven fabric and knitted fabric after multiple household washing, and the shrinkage of various textiles after washing by conventional household washing method is detected.
Place the specimen on the measuring table and make at least three pairs of marks on it in both length and width directions. Ensure that the distance between the marks of each pair is at least 350mm, that no mark is less than 50mm from the edges of the specimen and that the measuring points are regularly spaced across the specimen.
Lay the specimen flat on the smooth, flat surface and remove wrinkles without stretching the specimen. Place the ruler on the specimen, taking care to avoid distortion of the specimen. Record the distance between the pairs of marks to the nearest 1mm. Flatten the wrinkled specimen by the instrument so that the measurement does not cause deviation.
The change rate of the dimension in the direction of length and the width direction, taking the average value of each measurement.
Describe the laundry conditions used, including water level, water temperature, and laundry procedure.
Describe the dry conditions used, including drying procedures and temperature.
Explain the appearance change of the test sample after washing.
How to test shrinkage? Through this article, I think you might have got some information. If you want to know more, please contact us. As a professional supplier of textile instruments, we are committed to continually improve the user experience, TESTEX Textile Testing Equipment is a reliable choice for testing textile quality.[contact-form-7 404 "Not Found"]