Bally Flexing Tester TF116
Bally Flexing Tester, is designed for flexing resistance test to bending of leather and fabric. Flexometer complies with ISO 5402, ISO 32100, SATRA TM55, DIN 53351, BS-3144 ,and etc.
Bally Flexing Tester (Flexometer) is designed for flexing resistance test to bending or other types of failure at flexing creases.
The Flexometer is applicable to all flexible materials, in particular leathers, artificial leather below thickness of 3.0mm, and other coated fabrics, sheet materials, etc.
This Flexometer is designed to conduct the flexometer test method: test piece is folded with the surface to be tested inwards and clamped in an upper movable clamp and with the surface to be tested outwards in a lower fixed clamp. Movement of the upper clam caused a fold in the test piece to run along with it. The test piece is examined periodically for damage.
- Work stations 12
- Counter 1 ~ 999,999 times
- Fiexing angle 22°30´± 30´
- Frequency 100 ± 5 cpm
|220/110 V||50/60 Hz||4 A|
|Length:||970 mm||Width:||560 mm||Height:||620 mm|
|ISO 5402||ISO 32100||SATRA TM55||DIN 53351|
|BS-3144||BS EN 13512||JIS-K6545||CNS-7705|
What is a flexometer
Flexometer, also known as bally leather flexing tester, leather bending tester, is an international instrument used to evaluate the bending resistance of various light leather and its surface modifiers. It is mainly used to test the bending resistance of leather under continuous bending. So how to use a flexometer for a flexometer test? The flexometer uses one side of the specimen as the internal test and the other side as the outer side for reciprocating bending to the damage of the test piece and record the bending times, or after bending a certain number of times and observe the degree of damage. Put the sample into the fixture of the testing machine, twists and turns, then observe the surface of the specimen with a magnifier, and record the degree of damage to test the flexure strength of the specimen.
Flexometer definition: the bending resistance of various leather under continuous bending, it is an important indicator of the durability of leather. We will provide you more flexometer description if you contact us.
Q: Is this machine to determine resistance of a coated fabric to repeated flexing under specific conditions?
A: Yes, it is.