Spray Rating Tester TF160
Spray Rating Tester, to determine the surface wetting resistance of fabrics, which may or may not have been given a water-resistant or water-repellent finish. Spray Rating Tester complies with AATCC 22, ISO 4920, EN24920, etc. Our Spray Tester is your best choice when you do spray test.
Spray Rating Tester, to determine the surface wetting resistance of fabrics, which may or may not have been given a water-resistant or water-repellent finish.
Spray rating tester consists of a metal framework allowing distilled water to be sprayed through a nozzle onto a test specimen at 45° and 150mm below the nozzle.
- Nozzle 19 holes of Di 0.9mm
- Angle of holder 45°
- Holder 150mm, 150mm below the nozzle
- Measuring Cup 250m
|Width:||260 mm||Length:||220 mm||Height:||550 mm|
|ISO 4920||AATCC 22||BS EN 24920||BS 3702|
|M/S P23||M/S P133||NEXT 23|
Spray Rating Test Method
Test piece preparation:
Test piece (Size: 18CM*18CM)
- As AATCC 22 required, we put the test piece to the humidifying treatment more than 4 hours, which under the environment of standard temperature of21±1℃and relative humility of 65±2℃.
- After the humidifying treatment, the test piece clamped by the holder on the support at a 45°, and we pure 50ml water into the funnel rapidly steadily in order to keep drenching constantly. When the water is used up, we can observe the test piece.
- According to description of spray rating to assess its grade of wet but without the intergrade.
Grade 1: All of the surface is wet.
Grade 2: Half of the surface is wet. It usually means the sum of small and disconnected areas of wet.
Grade3: Only small and disconnected areas of wet on the surface.
Grade4: The surface appears water drop but not wet.
Grade5: The surface is neither wet nor water drop.
The test result should include the following message:
- The illustration of the test procedure according to the standard.
- Atmospheric conditions.
- Water temperature.
- Spray rating of each test piece.
- Experimental date.