Vertical Flammability Chamber TF312
Vertical Flammability Chamber for measuring the vertical flame spread of children’s sleepwear, fabrics (fabric burn test), other textile materials or resilient filling materials used in upholstered furniture. Vertical Flammability Chamber complies with ASTM D6413, CALIF TB117, 16CFR Parts1615/1616, etc. Please contact us for more information about flammability test method.
Vertical Flammability Chamber, for measuring the vertical flame spread of children’s sleepwear, fabrics (fabric burn test), and other textile materials or resilient filling materials used in upholstered furniture.
The burner ignites the specimen automatically, and ignition time is programmable; after-flame time and afterglow time are timed automatically and shown on the display. The user-friendly design of the control panel, Touch screen is equipped to make the operation conveniently, the test results are displayed on the screen after the vertical flammability test.
Comprises a draft-free stainless steel flammability chamber with observation window for easy test viewing. Specimen holders, burner and associated hardware are dependent on the test standard and must be specified.
- System and Display Programmable PLC and touch screen controls Time Display and accuracy 0-999.9s, 0.1s
- Ignition time 12+/-2 s or 3+/-0.2 s
- Burner size 10mm / 11mm x barrel length 76+/-6mm
- burner to specimen 19 or 17 mm
- Dimension of holder outside 442 x 76 / 442 x 89mm, interior 356×51 mm
- Timing PLC controlled
Optional Testing Kits
- FTMS 191-5903 (fabric)
- DOC FF3/71 Children pyjamas / CPAI 84 Camping tentage material
- CALIF TB-117 Resilient Filling Materials Vertical
|220/110 V||50/60 Hz||2 A|
|Width:||570 mm||Depth:||380 mm||Height:||850 mm|
|23 inch||15 inch||34 inch|
|ASTM D6413||CALIF TB117||DOC-FF 3||DOC-FF 71|
|16CFR Parts 1615||16CFR Parts 1616||FTMS 191-5903||CPAI 84|
What is the Vertical Flammability Chamber?
The vertical flammability tester is used to test the combustion performance of children’s pajamas or decorative fabrics. Sealed stainless steel combustor, an observation window and a standard burner, which can test flame spread speed accurately. With different sample clip and accessory test package to meet different test standards. Control the testing process easily through an automatic gas control system. Here take the ASTM D6413 standard as an example to introduce the vertical flammability test of the fabric.
Adjust the gas pressure to 17.2±1.7KPa and ignite the flame, adjust the flame height to 3mm, and ensure that the shape of the flame will not change in 12s. Close the vent and open the solenoid valve to ensure that the flame height reaches 38mm. Adjust the burner position so that the lower edge of the sample clip is in the middle of 19mm at the top of the burner. Adjust the timer to ensure that the ignite time of the sample is 12±0.2s.
Cut 5 samples along the longitudinal and transverse directions, the size is 76mm * 300mm, and ensure that the sample is flat. Then adjust the humidity of the samples as required and store them in the dryer.
1 Remove the sample from the dryer, install the sample and set it on fire in 4 minutes.
2 Fix the sample with the sample clip, then put it into the test box, and adjust the burner so that the center point of the bottom edge of the sample is located at the position of 19mm above the burner.
3 Open the igniter and ignite the sample for 12±0.2s. Observe the melting and dropping of the sample during combustion, and record the observation results.
4 After removing the fire source, record the time of continuous combustion and the flameless combustion time of the material. Don’t extinguish Mars because it affects the carbonation length.
5 Remove the sample clip from the test box and open the ventilation cabinet to remove the smoke from the test box.
6 Measure the carbonation length along the intact edge of the sample.
What should be Paying Attention to in Vertical Burning Testing?
1 Whether the sample and the sample rack are accurately installed is an important factor that affects the vertical burning test.
2 When cutting the sample, you should try to cut flat and wrinkle-free samples. The upper and lower ends of the sample should be trimmed flush, the size should be as accurate as possible.
3 When placing the specimen on the specimen holder, care should be taken so that the lower end of the specimen is flush with the lower end of the specimen holder to ensure accurate measurement of the length during the vertical burning test.
4 Suspension of the sample holder should be hanging vertically above the igniter, not tilted. The center of the sample should be in the flame.